|Publication number||USRE39977 E1|
|Application number||US 11/103,423|
|Publication date||1 Jan 2008|
|Filing date||11 Apr 2005|
|Priority date||13 Oct 2000|
|Also published as||US6734962, US7019296, US7061606, US7068357, US7123360, US7268861, US7268862, US7317516, US7436500, US20020113210, US20040159789, US20060033026, US20060049354, US20060091311, US20060151702, US20060157652, US20060164640, US20060192956|
|Publication number||103423, 11103423, US RE39977 E1, US RE39977E1, US-E1-RE39977, USRE39977 E1, USRE39977E1|
|Inventors||Patrick J. Treado, Matthew Nelson, Scott Keitzer|
|Original Assignee||Chemimage Corporation|
|Export Citation||BiBTeX, EndNote, RefMan|
|Patent Citations (51), Non-Patent Citations (62), Referenced by (11), Classifications (17), Legal Events (4)|
|External Links: USPTO, USPTO Assignment, Espacenet|
This application claims the benefit of U.S. Provisional Application No. 60/239,969, entitled “Near Infrared Chemical Imaging Microscope” filed Oct. 13, 2000.
This work is supported by the National Institute of Standards and Technology (NIST) under the Advanced Technology Program (ATP) award (Contract Number 70NANB8H4021)
The present invention is related to near-infrared (NIR) microscopes for spectroscopic and image analysis, and, in particular, to microscopes useful for both NIR spectroscopy, NIR chemical imaging and NIR volumetric chemical imaging.
NIR spectroscopy is a mature, non-contact, non-destructive analytical characterization tool that has been widely applied to a broad range of materials. The NIR region of the electromagnetic spectrum encompasses radiation with wavelength of 0.78 to 2.5 μm (12,800 to 4,000 cm−1). NIR spectra result from the overtone and combination bands of fundamental mid-infrared (MIR) bands. Among the many desirable characteristics, NIR is used to rapidly obtain both qualitative and quantitative information about the molecular makeup of a material. Digital imaging, on the other hand, provides a means to obtain optical (i.e., spatial—morphological, topographical, etc.) information about a material. By combining the spatial information obtained from digital imagery and the spectral information obtained from NIR spectroscopy, the chemical makeup of complex material matrices can be mapped out in both two and three spatial dimensions. NIR chemical imaging combines NIR spectroscopy and digital imaging for the molecular-specific analysis of materials. A NIR chemical imaging microscope apparatus employing NIR absorption molecular spectroscopy for materials characterization is disclosed.
NIR microscopes are used to obtain NIR absorption, transmittance or reflectance spectra (e.g., NIR microspectra) from samples ranging in size between 1 and 1000 μm. These instruments are typically equipped with a digital camera to visually locate a region of interest on a sample upon which a NIR light beam from a Fourier transform (FT) spectrometer is focused. Reflective optics are used to direct the transmitted or reflected light from the sample to a NIR detector. The output is a NIR absorption spectrum collected in transmittance or reflectance mode.
NIR chemical imaging can be considered an extension of NIR microspectroscopy. Much of the imaging performed since the development of the first NIR microprobes has involved spatial scanning of samples beneath NIR microscopes in order to construct NIR “maps” of surfaces. In point by point scanning with NIR microscopes, the NIR light beam is focused onto the surface of a sample or apertured to illuminate a small region of a sample and a spectrum from each spatial position is collected. Images are obtained by rastering the sample through the focused or apertured NIR light beam and the spectra recorded are then reconstructed to form an image. Although point scanning produces images based on NIR contrast, long experimental times are common since the duration of the experiment is proportional to the number of image pixels. As a direct result, point scan images are captured at low image definition, which relates directly to the limited utility of the technique as an imaging tool for the routine assessment of material morphology. The spatial resolution of the image is limited by the size of the NIR illumination spot on the sample (no less than 1 μm) and the rastering mechanism, which requires the use of moving mechanical parts that are challenging to operate reproducibly.
NIR imaging cameras have been used in photography for decades. Until recently, however, it has not been easily accessible to those not versed in traditional photographic processes. By using optical filters (e.g., cold filters) that block the visible wavelengths (0.4-0.78 μm), charge-coupled devices (CCDs) used in digital cameras and camcorders can be used to sense NIR light out to around 1100 nm. Other regions of the NIR spectrum can be viewed using devices such as indium gallium arsenide (InGaAs—0.9 μm to 1.7 μm) and indium antimonide (InSb—1.0 μm to 5.0 μm) focal plane array (FPA) detectors. These integrated wavelength NIR imaging approaches allow one to study relative light intensities of objects over broad ranges of the NIR spectrum, but useful chemical information is unattainable without the use of some type of discrete wavelength filtering device.
The use of dielectric interference filters in combination with NIR FPAs is one method in which chemical information can be obtained from a sample. To form NIR chemical images, a NIR light beam is defocused to illuminate a wide field of view and the reflected or transmitted light from the illuminated area is imaged onto a two-dimensional NIR detector. A selection of discrete dielectric interference filters provided in a filter wheel, or a linearly variable or circularly variable format can be positioned in front of a broadband NIR light source, or in front of the NIR FPA itself in order to collect NIR wavelength resolved images. Typically, the use of several fixed bandpass filters is required to access the entire NIR spectrum. The spatial resolution of the NIR image approaches that of the optical microscope, while spectral resolution of several nanometers has been demonstrated. Key limitations of the dielectric filter approach include the need for a multitude of discrete filters to provide appreciable free spectral range, or the reliance on moving mechanical parts in employing continuously tunable dielectric interference filters as a requirement to form wavelength resolved images. While moving mechanical assemblies can be engineered they add cost and complexity to NIR chemical imaging systems. Alternatives to moving mechanical assemblies are generally more cost effective and provide performance advantages.
Acoustic-optic tunable filters (AOTFs) have been employed as no-moving-parts imaging spectrometers for NIR imaging. The AOTF is a solid-state device that is capable of functioning from the UV to the mid-IR depending on the choice of the filter's crystal material. Operation of the AOTF is based on the interaction of light with a traveling acoustic sound wave in an anisotropic crystal medium. The incident light is diffracted with a narrow spectral bandpass when an rf signal is applied to the device. By changing the applied rf frequency under computer control the spectral passband can be tuned rapidly with the benefit of non-moving parts.
For use in NIR chemical imaging, AOTFs have distinct limitations. AOTFs have imaging performance that is degraded appreciably from diffraction-limited conditions due to dispersion effects and image shifting effects. Furthermore, AOTFs suffer from temperature instability and exhibit nonlinear properties that complicate their use as imaging spectrometers.
An aim of NIR chemical imaging technology development has been to develop a NIR imaging technique that combines diffraction-limited spatial resolution with high spectral resolution. NIR chemical imaging techniques have only recently achieved a degree of technological maturity that allow the collection of high resolution (spectral and spatial) data with the advent of the liquid crystal (LC) imaging spectrometers. In general, LC devices provide diffraction-limited spatial resolution. The spectral resolution of the LC imaging spectrometer is comparable to that provided by dispersive monochromator and Fourier transform interferometers. In addition, LC technology provides high out of band rejection, broad free spectral range, moderate transmittance, high overall etendue and highly reproducible random access computer controlled tuning.
Under normal NIR imaging operation, LC imaging spectrometers allow NIR chemical images of samples to be recorded at discrete wavelengths (energies). A spectrum is generated corresponding to thousands of spatial locations at the sample surface by tuning the LC imaging spectrometer over a range of wavelengths and collecting NIR images systematically. Contrast is generated in the images based on the relative amounts of NIR absorption, transmittance or reflectance that is generated by the different species located throughout the sample. Since a high quality NIR spectrum is generated for each pixel location, a wide variety of chemometric analysis tools, both univariate and multivariate, can be applied to the NIR image data to extract pertinent information. Correlative multivariate routines are particularly powerful when applied to chemical images collected from samples intentionally seeded with a known standard material. This approach of incorporating calibration standards within an image field of view can be extended to quantitative chemical image analysis. In addition, digital image analysis procedures can also be applied to high image quality NIR chemical images to perform routine particle analysis in both two (2D) and three (3D) spatial dimensions. Volumetric 3D NIR chemical image and analysis can be performed very effectively using numerical deconvolution computational strategies.
To address the need for a device that can provide video imaging, NIR spectroscopy and high resolution (spatial and spectral) NIR chemical imaging in two and three spatial dimensions, a novel NIR chemical imaging microscope has been developed that is NIR chemical imaging capable.
The microscope design uses NIR optimized liquid crystal (LC) imaging spectrometer technology for wavelength selection. The NIR optimized refractive microscope is used in conjunction with infinity-corrected objectives to form the NIR image on the detector with or without the use of a tube lens. An integrated parfocal analog color CCD detector provides real-time sample positioning and focusing. The color image and the NIR image are fused in software. In one configuration, the NIR microscope may be used as a volumetric imaging instrument through the means of moving the sample through focus, collecting images at varying focal depths and reconstructing a volumetric image of the sample in software, or through the means of keeping the sample fixed and changing the wavelength dependent depth of penetration in conjunction with a refractive tube lens with a well characterized chromatic effect. The output of the microscope can be coupled to a NIR spectrometer either via direct optical coupling or via a fiber optic. A Chemical Imaging Addition Method seeds the sample with a material of known composition, structure and/or concentration and then generates the NIR image suitable for qualitative and quantitative analysis. The microscope generates NIR chemical image data that is analyzed and visualized using chemical image analysis software in a systematic and comprehensive manner. While this invention has been demonstrated on a microscope optic platform, the novel concepts are also applicable to other image gathering platforms, namely fiberscopes, macrolens systems and telescopes.
The NIR chemical imaging microscope combines in a single platform a NIR optimized refractive optical microscope base, which is equipped with NIR optimized infinity-corrected microscope objectives, an automated XYZ translational microscope stage and quartz tungsten halogen (QTH) lamps to secure and illuminate samples for NIR spectroscopy and imaging, an analog color charge-coupled device (CCD) detector for ordinary optical image collection and digital image collection, a NIR LC imaging spectrometer for NIR chemical image wavelength selection and a room temperature or optionally cooled NIR FPA for NIR image capture.
Ordinary optical imagery of the sample can be obtained using a mirror or beamsplitter or prism arrangement inserted into turret 6 and collecting an image with an analog or digital color or monochrome charge-coupled device (CCD) or CMOS detector 7. In NIR chemical imaging mode, the magnified NIR image is coupled through a NIR LC imaging spectrometer 8 and collected on a room temperature or cooled NIR focal plane array (FPA) detector 9. The FPA is typically comprised of indium gallium arsenide (InGaAs), but may be comprised of other NIR sensitive materials, including platinum silicide (PtSi), indium antimonide (InSb) or mercury cadmium telluride (HgCdTe). Using a beam-splitting element inserted into turret 6, NIR and ordinary optical imagery can be collected with an analog monochrome or color CCD detector 7 and NIR FPA 9 simultaneously.
A central processing unit 10, typically a Pentium computer, is used for NIR chemical image collection and processing. The analog color CCD 7, NIR FPA 9, automated XYZ translational microscope stage 4 controlled via a controller 12 and NIR LC imaging spectrometer 8 (through LC imaging spectrometer controller 11) are operated with commercial software, such as Acquisition Manager (ChemIcon Inc.) in conjunction with ChemImage (ChemIcon Inc.).
By introducing a polarization sensitive beam splitting element in the optical path prior to the NIR LC imaging spectrometer 8 (not shown in schematic diagram), a portion of the NIR light from the sample may be coupled to a remote NIR spectrometer (also not shown in schematic diagram).
Preferably, NIR optimized liquid crystal (LC) imaging spectrometer technology is used for wavelength selection. The LC imaging spectrometer may be of the following types: Lyot liquid crystal tunable filter (LCTF); Evans Split-Element LCTF; Solc LCTF; Ferroelectric LCTF; Liquid crystal Fabry Perot (LCFP); or a hybrid filter technology comprised of a combination of the above-mentioned LC filter types or the above mentioned filter types in combination with fixed bandbass and bandreject filters comprised of dielectric, rugate, holographic, color absorption, acousto-optic or polarization types.
One novel component of this invention, is that a NIR optimized refractive microscope is used in conjunction with infinity-corrected objectives to form the NIR image on the detector without the use of a tube lens. The microscope can be optimized for NIR operation through inherent design of objective and associated anti-reflective coatings, condenser and light source. To simultaneously provide high numerical apertures the objective should be refractive. To minimize chromatic aberration, maximize throughput and reduce cost the conventional tube lens can be eliminated, while having the NIR objective form the NIR image directly onto the NIR focal plane array (FPA) detector, typically of the InGaAs type. The FPA can also be comprised of Si, SiGe, PtSi, InSb, HgCdTe, PdSi, Ge, analog vidicon types. The FPA output is digitized using an analog or digital frame grabber approach.
An integrated parfocal analog CCD detector provides real-time sample positioning and focusing. An analog video camera sensitive to visible radiation, typically a color or monochrome CCD detector, but may be comprised of a CMOS type, is positioned parfocal with the NIR FPA detector to facilitate sample positioning and focusing without requiring direct viewing of the sample through conventional eyepieces. The video camera output is typically digitized using a frame grabber approach.
The color image and the NIR image are fused using software. While the NIR and visible cameras often generate images having different contrast, the sample fields of view can be matched through a combination of optical and software manipulations. As a result, the NIR and visible images can be compared and even fused through the use of overlay techniques and correlation techniques to provide the user a near-real time view of both detector outputs on the same computer display. The comparative and integrated views of the sample can significantly enhance the understanding of sample morphology and architecture. By comparing the visible, NIR and NIR chemical images, additional useful information can be acquired about the chemical composition, structure and concentration of species in samples.
The NIR microscope can be used as a volumetric imaging instrument through the means of moving the sample through focus in the Z, axial dimension, collecting images in and out of focus and reconstructing a volumetric image of the sample in software. For samples having some volume (bulk materials, surfaces, interfaces, interphases), volumetric chemical imaging in the NIR has been shown to be useful for failure analysis, product development and routine quality monitoring. The potential also exists for performing quantitative analysis simultaneous with volumetric analysis. Volumetric imaging can be performed in a non-contact mode without modifying the sample through the use of numerical confocal techniques, which require that the sample be imaged at discrete focal planes. The resulting images are processed and reconstructed and visualized. Computational optical sectioning reconstruction techniques based on a variety of strategies have been demonstrated, including nearest neighbors and iterative deconvolution.
An alternative to sample positioning combined with computation reconstruction is to employ a tube lens in the image formation path of the microscope which introduces chromatic aberration. As a result the sample can be interrogated as a function of sample depth by exercising the LC imaging spectrometer, collecting images at different wavelengths which penetrate to differing degrees into bulk materials. These wavelength dependent, depth dependent images can be reconstructed to form volumetric images of materials without requiring the sample to be moved, again through application of computational optical sectioning reconstruction algorithms.
The output of the microscope can be coupled to a NIR spectrometer either via direct optical coupling or via a fiber optic cable. This allows conventional spectroscopic tools to be used to gather NIR spectra for traditional, high speed spectral analysis. The spectrometers can be of the following types: fixed filter spectrometers; grating based spectrometers; Fourier Transform spectrometers; or Acousto-Optic spectrometers.
A novel method that is readily employed by the disclosed microscope invention is a method described as the Chemical Imaging Addition Method which involves seeding the sample with a material of known composition, structure and/or concentration and then generating the NIR image suitable for qualitative and quantitative analysis. The Chemical Imaging Addition Method is a novel extension of a standard analytical chemical analysis technique, the Standard Addition Method. A common practice in quantitative chemical analysis is to construct a standard calibration curve which is a plot of analytical response for a particular technique as a function of known analyte concentration. By measuring the analytical response from an unknown sample, an estimate of the analyte concentration can then be extrapolated from the calibration curve. In the Standard Addition Method, known quantities of the analyte are added to the samples and the increase in analytical response is measured. When the analytical response is linearly related to concentration, the concentration of the unknown analyte can be found by plotting the analytical response from a series of standards and extrapolating the unknown concentration from the curve. In this graph, however, the x-axis is the concentration of added analyte after being mixed with the sample. The x-intercept of the curve is the concentration of the unknown following dilution. The primary advantage of the standard addition method is that the matrix remains constant for all samples.
While the Standard Addition Method is used specifically for quantitative analysis, the Chemical Imaging Addition Method can be used for qualitative and quantitative analysis. The Chemical Imaging Addition Method relies upon spatially isolating analyte standards in order to calibrate the Chemical Imaging analysis. In chemical imaging, thousands of linearly independent, spatially-resolved spectra are collected in parallel of analytes found within complex host matrices. These spectra can then be processed to generate unique contrast intrinsic to analyte species without the use of stains, dyes, or contrast agents. Various spectroscopic methods including near-infrared (NIR) absorption spectroscopy can be used to probe molecular composition and structure without being destructive to the sample. Similarly, in NIR chemical imaging the contrast that is generated reveals the spatial distribution of properties revealed in the underlying NIR spectra.
The Chemical Imaging Addition Method can involve several data processing steps, typically including, but not limited to:
Effective materials characterization with the disclosed NIR chemical imaging microscope invention typically requires application of a multitude of software procedures to the NIR chemical image. A schematic of the chemical image analysis cycle is shown in
Until recently, seamless integration of spectral analysis, chemometric analysis and digital image analysis has not been commercially available. Individual communities have independently developed advanced software applicable to their specific requirements. For example, digital imaging software packages that treat single-frame gray-scale images and spectral processing programs that apply chemometric techniques have both reached a relatively mature state. One limitation to the development of chemical imaging, however, has been the lack of integrated software that combines enough of the features of each of these individual disciplines to have practical utility.
Historically, practitioners of chemical imaging were forced to develop their own software routines to perform each of the key steps of the data analysis. Typically, routines were prototyped using packages that supported scripting capability, such as Matlab, IDL, Grams or LabView. These packages, while flexible, are limited by steep learning curves, computational inefficiencies, and the need for individual practitioners to develop their own graphical user interface (GUI). Today, commercially available software does exist that provides efficient data processing and the ease of use of a simple GUI.
Software that meets these goals must address the entirety of the chemical imaging process. The chemical imaging analysis cycle illustrates the steps needed to successfully extract information from chemical images and to tap the full potential provided by chemical imaging systems. The cycle begins with the selection of sample measurement strategies and continues through to the presentation of a measurement solution. The first step is the collection of images. The related software must accommodate the full complement of chemical image acquisition configurations, including support of various spectroscopic techniques, the associated spectrometers and imaging detectors, and the sampling flexibility required by differing sample sizes and collection times. Ideally, even relatively disparate instrument designs can have one intuitive GUI to facilitate ease of use and ease of adoption.
The second step in the analysis cycle is data preprocessing. In general, preprocessing steps attempt to minimize contributions from chemical imaging instrument response that are not related to variations in the chemical composition of the imaged sample. Some of the functionalities needed include: correction for detector response, including variations in detector quantum efficiency, bad detector pixels and cosmic events; variation in source illumination intensity across the sample; and gross differentiation between spectral lineshapes based on baseline fitting and subtraction. Examples of tools available for preprocessing include ratiometric correction of detector pixel response; spectral operations such as Fourier filters and other spectral filters, normalization, mean centering, baseline correction, and smoothing; spatial operations such as cosmic filtering, low-pass filters, high-pass filters, and a number of other spatial filters.
Once instrument response has been suppressed, qualitative processing can be employed. Qualitative chemical image analysis attempts to address a simple question, “What is present and how is it distributed?”. Many chemometric tools fall under this category. A partial list includes: correlation techniques such as cosine correlation and Euclidean distance correlation; classification techniques such as principal components analysis, cluster analysis, discriminant analysis, and multi-way analysis; and spectral deconvolution techniques such as SIMPLISMA, linear spectral unmixing and multivariate curve resolution.
Quantitative analysis deals with the development of concentration map images. Just as in quantitative spectral analysis, a number of multivariate chemometric techniques can be used to build the calibration models. In applying quantitative chemical imaging, all of the challenges experienced in non-imaging spectral, analysis are present in quantitative chemical imaging, such as the selection of the calibration set and the verification of the model. However, in chemical imaging additional challenges exist, such as variations in sample thickness and the variability of multiple detector elements, to name a few. Depending on the quality of the models developed, the results can range from semiquantitative concentration maps to rigorous quantitative measurements.
Results obtained from preprocessing, qualitative analysis and quantitative analysis must be visualized. Software tools must provide scaling, automapping, pseudo-color image representation, surface maps, volumetric representation, and multiple modes of presentation such as single image frame views, montage views, and animation of multidimensional chemical images, as well as a variety of digital image analysis algorithms for look up table (LUT) manipulation and contrast enhancement.
Once digital chemical images have been generated, traditional digital image analysis can be applied. For example, Spatial Analysis and Chemical Image Measurement involve binarization of the high bit depth (typically 32 bits/pixel) chemical image using threshold and segmentation strategies. Once binary images have been generated, analysis tools can examine a number of image domain features such as size, location, alignment, shape factors, domain count, domain density, and classification of domains based on any of the selected features. Results of these calculations can be used to develop key quantitative image parameters that can be used to characterize materials.
The final category of tools, Automated Image Processing, involves the automation of key steps or of the entire chemical image analysis process. For example, the detection of well defined features in an image can be completely automated and the results of these automated analyses can be tabulated based on any number of criteria (particle size, shape, chemical composition, etc). Automated chemical imaging platforms have been developed that can run for hours in an unsupervised fashion.
This invention incorporates a comprehensive analysis approach that allows user's to carefully plan experiments and optimize instrument parameters and should allow the maximum amount of information to be extracted from chemical images so that the user can make intelligent decisions.
As the demand for high quality, low cost X-ray, γ-ray and imaging detector devices increases, there is a need to improve the quality and production yield of semiconductor materials used in these devices. One effective strategy for improving semiconductor device yield is through the use of better device characterization tools that can rapidly and nondestructively identify defects at early stages in the fabrication process. Early screening helps to elucidate the underlying causes of defects and to reduce downstream costs associated with processing defect laden materials that are ultimately scrapped. The present invention can be used to characterize tellurium inclusion defects in cadmium zinc telluride (CdZnTe) semiconductor materials based on near infrared imaging. With this approach, large area wafers can be inspected rapidly and non-destructively in two and three spatial dimensions by collecting NIR image frames at multiple regions of interest throughout the wafer using an automated NIR imaging system. The NIR image frames are subjected to image processing algorithms including background correction and image binarization. Particle analysis is performed on the binarized images to reveal tellurium inclusion statistics, sufficient to pass or fail wafers. In addition, data visualization software is used to view the tellurium inclusions in two and three spatial dimensions.
The present invention has been used to automatically inspect tellurium inclusions in CdZnTe. Compound semiconductors are challenging to fabricate. There are several steps along the manufacturing process in which defects can arise. The chemical nature associated with semiconductor defects often plays a vital role in device performance. Device fabrication and device processing defects can be difficult and time consuming to measure during manufacturing. Unfortunately, defective devices are often left undiagnosed until latter stages in the manufacturing process because of the inadequacy of the metrology tools being used. This results in low production yields and high costs which can be an impediment to growth in the semiconductor device market potential.
There is a general need in the semiconductor industry for metrology technologies that can nondestructively assess semiconductor material defects and ultimately increase manufacturing yields. A potential solution is to develop a high throughput screening system capable of fusing multiple chemical imaging modalities into a single instrument. Chemical imaging combines digital imaging and molecular spectroscopy for the chemical analysis of materials. A modality of based on near-infrared (NIR) chemical imaging can be used to inspect tellurium inclusions in CdZnTe compound semiconductor materials.
CdZnTe is a leading material for use in room temperature X-ray detectors, γ-ray radiation detectors and imaging devices. Applications for these devices include nuclear diagnostics, digital radiography, high-resolution astrophysical X-ray and γ-ray imaging, industrial web gauging and nuclear nonproliferation. These devices are often decorated with microscopic and macroscopic defects limiting the yield of large-size, high-quality materials. Defects commonly found in these materials include cracks, grain boundaries, twin boundaries, pipes, precipitates and inclusions. CdZnTe wafers are often graded based on the size and number of Te inclusion defects present.
The definition used by Rudolph and Muhlberg for tellurium inclusions (i.e., tellurium-rich domains in the 1-50 μm size range that originate as a result of morphological instabilities at the growth interface as tellurium-rich melt droplets are captured from the boundary layer ahead of the interface) has been adopted and is used herein. There have been numerous studies on the composition and distribution of tellurium inclusions in CdZnTe material. It has been demonstrated that the presence of tellurium inclusions can impair the electronic properties of CdZnTe materials—consequently degrading the end-product device performance.
The current procedure used by low volume semiconductor manufacturers for characterizing tellurium inclusions in CdZnTe is labor intensive, susceptible to human error and provides little information on inclusions in the 1-5 μm size scale. Inclusions are viewed and counted manually by a human operator using an IR microscope platform. When an inclusion is identified that is suspected to exceed a specified size limit, a Polaroid film photograph is taken. An overlay of a stage micrometer is laid over the photograph to determine the size. This analysis is relatively time consuming, often taking several minutes to characterize a region of interest from a large wafer.
The present invention can be used for automated characterization of microscale tellurium inclusions in CdZnTe based on volumetric NIR chemical imaging. The system takes advantage of the fact that CdZnTe is transparent to infrared wavelengths (>850 nm). When viewing CdZnTe with an infrared focal plane array (IR-FPA) through a NIR LC imaging spectrometer, tellurium inclusions appear as dark, absorbing domains. The invention images wafers in two and three spatial dimensions capturing raw infrared images at each region of interest. Images are automatically background equilibrated, binarized and processed. The processed data provides particle statistical information such as inclusion counts, sizes, density, area and shape. The system provides a rapid method for characterizing tellurium inclusions as small as 0.5 μm while virtually eliminating the subjectivity associated with manual inspection.
Tellurium-rich CdZnTe samples were produced by a commercial supplier (eV Products) for analysis. Samples containing high tellurium inclusion densities were purposely acquired to effectively demonstrate the capabilities of the automated tellurium inclusions mapping system. The CdZnTe materials were grown by the Horizontal Bridgeman (HB) method and contained a nominal zinc cation loading concentration of 4% and an average etch pit density of 4×104/cm2. The materials displayed a face A <111> orientation and were polished on both sides. Sample thicknesses ranged from approximately 1 mm to 15 mm. No further sample preparation was necessary for the automated tellurium inclusion mapping analysis.
Volumetric maps of the tellurium inclusions in the CdZnTe samples were obtained by first placing the sample on the XYZ-translational stage of the automated mapping system. NIR image frames were then captured through the LC imaging spectrometer at a wavelength that maximized the Te precipitate contrast relative to the surrounding CdZnTe matrix in the X-Y direction at multiple regions of interest across the samples. Depth profiling was achieved by translating the sample focus under the microscope at user-defined increments. This process was then repeated in an iterative fashion until the entire wafer was characterized.
Once imaging data was collected, ChemImage was used to process the data. For each wafer, the software generates a background-corrected grayscale image, a binarized image using the threshold value selected for each frame of the image, a montage view of the binarized image and particle statistics. The particle statistics table includes information such as particle counts, particle sizes, particles densities, and a number of geometrical parameters such as particle area and particle aspect ratios.
Background Correction and Image Binarization
The automated particle analysis begins by applying a background correction preprocessing routine to the raw image frames. One of the biggest problems with the raw images collected is the gradually varying background across each image frame. As a result, a particle in one area of a frame may have a higher intensity value than the background of another area of that frame.
To address this issue, a background correction step is used to force the background to be essentially constant across a given image frame. The procedure applies a moving window across the image frame and smoothes the resulting background before subtracting it from the frame. Other operations such as low pass filtering and selective removal of bad camera pixels are also applied.
The second step in the automated particle analysis is the selection of the threshold value resulting in the binarized image which best reflects the number and size of particles actually present in the sample being imaged. A human operator would typically approach this problem by trying multiple threshold values and comparing the resulting binarized images to the actual image to see which binarized image best matches their perception of the particles in the actual image. The algorithm employed by the NIR chemical imaging microscope system takes essentially the same approach. A series of threshold values are used to generate binarized images. Each binarized image is submitted to a routine that finds the particles present in the image. A set of particle morphology rules was developed to determine the point at which the threshold value identifies the particles consistent with results obtained by a trained human operator. This threshold value is then further refined with using derivative operations.
Volumetric Reconstruction and Visualization
It is of particular interest to the semiconductor manufacturing industry to view defects, including tellurium inclusions in this example, in a three dimensional volumetric view. Individual binarized image frames generated at discrete axial planes of focus have been reconstructed into a volumetric view allowing users to view tellurium inclusions in three-dimensional space.
Table 1 provides tabulated statistical information on the volumetric data shown in FIG. 7.
Slice Number and Depth (μm)
# of Inclusions
Mean Diameter (μm)
Maximum Chord Length
Feret 1 Diameter
Feret 2 Diameter
Defects such as tellurium inclusions affect the electrical properties in CdZnTe semiconductor materials, degrading end-product device performance. Having the ability to rapidly and non-invasively identify and quantify tellurium inclusion defects at critical stages in the fabrication process provides semiconductor manufacturers with information that will enable them to optimize the manufacturing process and reduce production costs. The Automated NIR Volumetric Mapping System described here is capable of providing such information. The system provides qualitative and quantitative information about tellurium inclusions present in CdZnTe wafers in two and three spatial dimensions. This system boasts improved spatial resolution (−0.5 μm) compared to systems currently used by many semiconductor manufacturers and it virtually eliminates the subjectivity associated with human counting and sizing measurements. Whole wafers are capable of being characterized in minutes.
While in the above example, the present invention has been demonstrated in connection with the characterization of semiconductors, it is to be expressly understood that the present invention can also be used in the characterization of other materials including, but not limited to, food and agricultural products, paper products, pharmaceutical materials, polymers, thin films and in medical uses.
Although present preferred embodiments of the invention have been shown and described, it should be distinctly understood that the invention is not limited thereto but may be variously embodied within the scope of the following claims.
|Cited Patent||Filing date||Publication date||Applicant||Title|
|US3806257||4 May 1973||23 Apr 1974||Corning Glass Works||Operator viewing optics for a slide classification system|
|US4802760||13 Jan 1988||7 Feb 1989||Mitsubishi Denki Kabushiki Kaisha||Raman microprobe apparatus for determining crystal orientation|
|US5194912||22 Dec 1989||16 Mar 1993||Renishaw Plc||Raman analysis apparatus|
|US5377003||29 Apr 1994||27 Dec 1994||The United States Of America As Represented By The Department Of Health And Human Services||Spectroscopic imaging device employing imaging quality spectral filters|
|US5377004||15 Oct 1993||27 Dec 1994||Kaiser Optical Systems||Remote optical measurement probe|
|US5381236||6 Feb 1992||10 Jan 1995||Oxford Sensor Technology Limited||Optical sensor for imaging an object|
|US5394499||15 Dec 1993||28 Feb 1995||Olympus Optical Co., Ltd.||Observation system with an endoscope|
|US5442438||13 Nov 1992||15 Aug 1995||Renishaw Plc||Spectroscopic apparatus and methods|
|US5493443||10 Dec 1993||20 Feb 1996||Bruker Analytisch Mebtechnik Gmbh||Lens for a FT-raman microscope|
|US5528393||12 Jul 1994||18 Jun 1996||Regents Of The University Of Colorado||Split-element liquid crystal tunable optical filter|
|US5623342||31 May 1996||22 Apr 1997||Renishaw Plc.||Raman microscope|
|US5626134||20 Apr 1995||6 May 1997||Zuckerman; Ralph||Method and apparatus for the measurement of analyte concentration levels by the steady-state determination of fluorescence lifetime|
|US5645550||8 Apr 1994||8 Jul 1997||Chiron Technolas Gmbh Ophthalmologische System||Method and apparatus for providing precise location of points on the eye|
|US5668664||24 May 1996||16 Sep 1997||Asahi Seimitsu Kabushiki Kaisha||Color separation prism assembly for C-mount camera|
|US5689333||10 May 1995||18 Nov 1997||Renishaw Plc||Spectroscopic apparatus and methods|
|US5710626||15 Nov 1996||20 Jan 1998||Westinghouse Savannah River Company||Rugged fiber optic probe for raman measurement|
|US5862273||21 Feb 1997||19 Jan 1999||Kaiser Optical Systems, Inc.||Fiber optic probe with integral optical filtering|
|US5866430||27 May 1997||2 Feb 1999||Grow; Ann E.||Raman optrode processes and devices for detection of chemicals and microorganisms|
|US5900975||30 Oct 1997||4 May 1999||Cognex Corporation||Ghost image extinction in an active range sensor|
|US5901261||19 Jun 1997||4 May 1999||Visionex, Inc.||Fiber optic interface for optical probes with enhanced photonic efficiency, light manipulation, and stray light rejection|
|US5910816||7 Jun 1996||8 Jun 1999||Stryker Corporation||Imaging system with independent processing of visible an infrared light energy|
|US5911017||28 Jul 1997||8 Jun 1999||Visionex, Inc.||Fiber optic interface for laser spectroscopic Raman probes|
|US5943122||10 Jul 1998||24 Aug 1999||Nanometrics Incorporated||Integrated optical measurement instruments|
|US5943129 *||7 Aug 1997||24 Aug 1999||Cambridge Research & Instrumentation Inc.||Fluorescence imaging system|
|US5974211||6 Feb 1998||26 Oct 1999||Kaiser Optical Systems||Enhanced collection efficiency fiber-optic probe|
|US6002476||22 Apr 1998||14 Dec 1999||Chemicon Inc.||Chemical imaging system|
|US6006001||2 Dec 1997||21 Dec 1999||The Research Foundation Of Cuny||Fiberoptic assembly useful in optical spectroscopy|
|US6052187||31 Aug 1998||18 Apr 2000||Containerless Research, Inc.||Hyperspectral polarization profiler for remote sensing|
|US6069690||13 Nov 1998||30 May 2000||Uniphase Corporation||Integrated laser imaging and spectral analysis system|
|US6088100||29 Jun 1998||11 Jul 2000||Massachusetts Institute Of Technology||Three-dimensional light absorption spectroscopic imaging|
|US6091872||28 Oct 1997||18 Jul 2000||Katoot; Mohammad W.||Optical fiber imaging system|
|US6175750||19 Mar 1999||16 Jan 2001||Cytometrics, Inc.||System and method for calibrating a reflection imaging spectrophotometer|
|US6181414||5 Aug 1999||30 Jan 2001||Morphometrix Technologies Inc||Infrared spectroscopy for medical imaging|
|US6222970||12 Mar 1999||24 Apr 2001||Cirrex Corp.||Methods and apparatus for filtering an optical fiber|
|US6274871||22 Oct 1998||14 Aug 2001||Vysis, Inc.||Method and system for performing infrared study on a biological sample|
|US6300618||11 Dec 1998||9 Oct 2001||Yokogawa Electric Corporation||High speed 3-dimensional confocal microscopic equipment|
|US6392752||12 Jun 2000||21 May 2002||Kenneth Carlisle Johnson||Phase-measuring microlens microscopy|
|US6403947 *||16 Mar 2000||11 Jun 2002||Cambridge Research & Instrumentation Inc.||High-efficiency multiple probe imaging system|
|US6404497||25 Jan 1999||11 Jun 2002||Massachusetts Institute Of Technology||Polarized light scattering spectroscopy of tissue|
|US6483641||27 Oct 1998||19 Nov 2002||Digital Optical Imaging Corporation||Apparatus and methods relating to spatially light modulated microscopy|
|US6485413||6 Mar 1998||26 Nov 2002||The General Hospital Corporation||Methods and apparatus for forward-directed optical scanning instruments|
|US6530882||30 Jun 2000||11 Mar 2003||Inner Vision Imaging, L.L.C.||Endoscope having microscopic and macroscopic magnification|
|US6571117||10 Aug 2001||27 May 2003||Ralf Marbach||Capillary sweet spot imaging for improving the tracking accuracy and SNR of noninvasive blood analysis methods|
|US6614532||28 Apr 2000||2 Sep 2003||Mcgill University||Apparatus and method for light profile microscopy|
|US6640130||3 Jul 2000||28 Oct 2003||Hypermed, Inc.||Integrated imaging apparatus|
|US6697665||11 Aug 1994||24 Feb 2004||Massachusetts Institute Of Technology||Systems and methods of molecular spectroscopy to provide for the diagnosis of tissue|
|US6711283 *||3 May 2000||23 Mar 2004||Aperio Technologies, Inc.||Fully automatic rapid microscope slide scanner|
|US20040004715||30 Jun 2003||8 Jan 2004||David Tuschel||Method for Raman imaging of semiconductor materials|
|EP0174778B1||29 Aug 1985||30 Nov 1988||Rank Cintel Limited||Apparatus for individually processing optical images of different wavelengths|
|WO1995011624A2||28 Oct 1994||4 May 1995||Feld Michael S||A raman endoscope|
|WO1998035262A1||6 Feb 1998||13 Aug 1998||Morphometrix Technologies Inc.||Infrared spectroscopy for medical imaging|
|1||"RAVEN Raman Imaging Fiberscope-A Tool for Insitu Material Analysis," Ben Franklin Technology Center of Western Pennsylvania, date unknown.|
|2||Christensen et al., "Raman Imaging Using a Tunable Dual-Stage Liquid Crystal Fabry-Perot Interferometer," Applied Spectroscopy, vol. 49, No. 8 (1995) pp. 1120-1125.|
|3||Christofides et al., "Reconstruction Mechanisms in Ion Implanted and Annealed Silicon Wafers," Defect and Diffusion Forum, vols. 117-118 (1985) pp. 45-64.|
|4||Dewilton et al., "A Raman Study of the Dopan Distribution in Submicron Pn Junctions in B+ or BF2+ Ion Implanted Silicon," SPIE vol. 623 Advanced Processing and Characterization of Semiconductors III (1986) pp. 26-34.|
|5||Dewilton et al., "Raman Spectroscopy For NonDestructive Depth Profile Studies Of Ion Implantation In Silicon", J. Electrochem. Soc.: Solid State Science And Technology, May 1986, pp. 988-993.|
|6||Dey et al., "Raman Scattering Characterization of Si(100) Implanted With Mega-Electron-Volt Sb", Journal of Applied Physics 87(3) Feb. 1, 2000, pp. 1110-1116.|
|7||Gerald C. Holst, "Electro-Optical Imaging System Performance," SPIE Optical Engineering Press, pp. 218-219, 238-239, 248-257,266-269, no date available.|
|8||Gift et al., "Near-Infrared Raman Imaging Microscope Based on Fiber-Bundle Image Compression," Journal of Raman Spectroscopy, vol. 30 (1999), pp. 757-765.|
|9||Gonzalez et al., "Digital Image Processing," Addison-Wesley Publishing Co. (1992) pp. 21-79.|
|10||Gorelick, "Raman and Non-Linear Light Scattering From Undersurface Layers Of Ion Implanted Silicon Crystals," Materials Science Forum, vol. 173-174 (1995) pp. 237-242.|
|11||H. Morris, C. Hoyt, P. Miller and P. Treado, "Liquid Crystal Tunalbe Filter Raman Chemical Imaging", vol. 50 Applied Spectroscopy, No. 6, pp. 805-811 (1996).|
|12||H. Skinner, T. Cooney , S. Sharma and S. Angel. "Remote Raman Microimaging Using an AOTF and a Spatially Coherent Microfiber Optical Probe". vol. 50 Applied Spectroscopy No. 8, pp. 1007-1014 (1996).|
|13||H.R. Morris and P.J. Treado, "LCTF Ramon Chemical Imaging of Thermoplastic Otefin (TPO) Architecture," Proc. Xvth ICORS, S.A. Asher, Ed (Willey, Chichester, 1996) 1186-1187.|
|14||I. Lewis and P. Griffiths, "Raman Spectrometry with Fiber-Optic Sampling", vol. 50 Applied Spectroscopy, No. 10, pp. 12A-29A (1996).|
|15||I. Lewis and P. Griffiths, "Raman Spectrometry with Fiber-Optic Sampling," Applied Spectroscopy, vol. 50, No. 10, (1996) pp. 12A-29A.|
|16||Ishioka et al., "Reduction in Raman Intensity of Si (1 1 1) Due to Defect Formation During Ion Irradiation," Solid State Communications, vol. 96, No. 6 (1995) pp. 387-390.|
|17||Jain et al., "Raman Scattering From Ion-Implanted Silicon," Physical Review B., vol. 32, No. 10, Nov. 15, 1985, pp. 6688-6691.|
|18||Jeff Hecht, "Imaging and Illuminating Fiber Optics," Chapter 28, (3rd ed. 1999) pp. 567-581.|
|19||John F. Turner ll and Patrick J. Treado, LCTF Raman Chemical Imaging in the Near-Infrared, Proc. SPIE 3061, (1997) 280-283.|
|20||Kirilov et al., "Amorphous Phase Transformation During Rapid Thermal Annealing of Ion-Implanted Si," Mat'l Res. Soc. Symp. Proc., vol. 52 (1986) pp. 131-138.|
|21||Lewis et al., "A Miniaturzed, No-Moving-Parts Raman Spectrometer," Applied Spectroscopy, vol. 47, No. 5, (1993) pp. 539-543.|
|22||M.D. Schaeberle and P.J. Treado, "LCTF Raman Chemical Imaging of Semiconductors," Proc. Xvth ICORS, S.A. Asher, Ed (Wiley, Chichester, 1996) 1188-1189.|
|23||Miller, Peter J. and Hoyt, Clifford C., "Multispectral Imaging with a Liquid Crystal Tunable Filter," SPIE vol. 2345, (1995) pp. 354-365.|
|24||Mizoguchi et al., "Raman Image Study of Flash-Lamp Annealing of Ion-Implanted Silicon," Journal of Applied Physics 77 (7) Apr. 1, 1995, pp. 3388-3392.|
|25||Morris et al., "Imaging Spectrometers for Fluorescence and Raman Microscopy: Acousto-Optic and Liquid Crystal Tunable Filter," Applied Spectroscopy, vol. 48, No. 7 (1994) pp. 857-866.|
|26||Morris, "Ultrasensitive Raman and Flourescence Imaging Using Liquid Crystal Tunable Filters," SPIE vol. 2385, (1995) pp. 81-87.|
|27||Morris, Hoyt, Miller and Treado, "Liquid Crystal Tunable Filter Raman Chemical Imaging," Applied Spectroscopy, No. 50, No. 6, Jun. 1996 pp. 805-811.|
|28||Nakashima et al., "Raman Microproble Study of Recrystallization in Ion-Implanted and Laser-Annealed Polycrystalline Silicon," Journal of Applied Physics 54(5) May 1983 pp. 2611-2617.|
|29||Othonos et al., "Multi-Wavelength Raman Probing of Phosphorus Implanted Silicon Wafers," Nucl. Instr. and Meth. in Phys. Rev. B., 117 (1996) pp. 367-374.|
|30||Othonos et al., "Multi-Wavelength Raman Probing of Phosphorus Implanted Silicon Wafers," Nuclear Instruments and Methods in Physics Research B., 117 (1996) pp. 367-374.|
|31||Othonos et al., "Raman Spectroscopy and Spreading Resistance Analysis of Phosphorous Implanted and Annealed Silicon," Journal of Applied Physics, 75 (12) Jun. 1994, pp. 8032-8038.|
|32||P. Treado et al., "High-Fidelity Raman Imaging Spectrometry: A Rapid Method Using an Acousto-Optic Tunable Filter". vol. 46 Applied Spectroscopy, No. 8, pp. 1211-(1992).|
|33||Patrick J. Treado and Michael D. Morris, Infrared and Raman Spectroscopic Imaging (Marcell Decker, New York, 1992) pp. 71-108.|
|34||Patrick J. Treado, "A Miniaturized Raman Fiberscope for Remote Chemical Imaging," Proposal Submitted to the Ben Franklin Technology Center of Western Pennsylvania, Mar. 21, 1, no date available.|
|35||Patrick J. Treado, "A Raman Endoscope for Breast Cancer Diagnosis," Proposal submitted to the Ben Franklin Technology Center of Western Pennsylvania, Apr. 8, 1998.|
|36||Patrick J. Treado, "Chemical Imaging Reveals More Than The Microscope," Laser Focus World, Oct., (1995) pp. 1-7.|
|37||Patrick J. Treado, Ira W. Levin and E. Neil Lewis, Near-Infrared Acousto-Optic Filtered Spectroscopic Microscopy: A Solid-State Approach to Chemical Imaging, Applied Spectroscopy 46, (1992) 553-559.|
|38||Patrick J. Treado, Ira W. Levin, and E. Neil Lewis, Indium Antimonide (InSb) Focal Plane Array (FPA) Detection for Near-Infrared Imaging Microscopy. Applied Spectroscopu 48. (1994) 607.|
|39||Patrick J. Treado, Letter to Peter Miller, Oct. 28, 1996.|
|40||Patrick J. Treado, Letter to Peter V. Foukal, Ph.D., Oct. 24, 1996.|
|41||Peter J. Miller, National Science Foundation SBIR Phase II Proposal, "High-Definition Raman Imaging Microscope," Oct. 1996.|
|42||Peter J. Miller, Small Business Innovation Research (SBIR) Phase I, "SBIR Phase I: High Definition Raman Imaging Microscope," Sep. 13, 1996.|
|43||Peter Miller, National Science Foundation Small Business Innovation Research Program, "High Definition Raman Imaging Microscope," Jun. 1994.|
|44||Robert D. Guenther, "Modern Optics, Polarization By BiRefringence," (John Wiley & Son 1990) pp. 529-534.|
|45||Schaeberle et al., "Multiplexed Acousto-Optic Tunable Filter (AOTF) Spectral Imaging Microscopy," SPIE, vol. 2173, (1994) pp. 11-20.|
|46||Schaeberle, et al., "Raman Chemical Imaging: Histopathology of Inclusions in Human Breast Tissue," Analytical Chemistry, vol. 68, No. 11, (1996), pp. 1829-1833.|
|47||Shukla et al., "A Raman Scattering From Ultraheavily-Iron Implanted and Laser-Annealed Silicon," Physical Review B. Vol. 34, No. 12, Dec. 15, 1986, pp. 8950-8953.|
|48||Skinner et al. "Remote Raman Microimaging Using An AOTF and a Spartially Coherent Microfiber Optical Probe," Applied Spectroscopy, vol. 50 No. 8, (1996) pp. 1007-1014.|
|49||Spectral Dimensions. NIR Systems Product Information, http://www.spectraldimensions.com/products/b-nir.html.|
|50||Treado et al, "Incorporation of a Band Pass Filter into a High Temperature Raman Illumination Fiberscope," Feb. 15, 1999.|
|51||Treado et al. "A Thousand Ponts of Light: The Hadamard Transform" Analytical Chemistry 61 (1989) Jun. 1, No. 11, pp 723-724.|
|52||Treado et al., "A Thousand Points Of Light: The Hadamard Transform In Chemical Analysis And Instrumentation," Analytical Chemistry, vol. 61, No. 11, Jun. 1, 1989, pp. 723-734.|
|53||Treado et al., "Engineering Study of the Feasibility of Incorporating a Raman Notch Filter into the Higher Temperature (HT) Fiberscope," Jul. 29, 1997.|
|54||Treado et al., "High-Fidelity Raman Imaging Spectrometry: A Rapid Method Using An Acousto-Optic Tunable Filter", Applied Spectroscopy, vol. 46, No. 8, (1992) pp. 1211-1216.|
|55||Treado et al., "Incorporation of a Notch Filter into A High Temperature Raman Collection Fiberscope," Jul. 20, 1998.|
|56||Treado et al., "Indium Antimonide (InSb) Focal Plane Array(FPA) Detection for Near-Infrared Imaging Microscopy," Applied Spectroscopy 48. (1994) p. 607.|
|57||Treado et al., "Indium Antimonide (InSb) Focal Plane Array(FPA) Detection for Near-Infrared Imaging Microscopy," Applied Spectroscopy vol. 48, No. 5. (1994) p. 607.|
|58||Treado et al., "Infrared Raman Spectroscopic Imaging," (Marcell Decker, 1992) pp. 71-108.|
|59||Treado et al., "Near-Infrared Acousto-Optic Filtered Spectroscopic Microscopy: A Solid-State Approach To Chemical Imaging", Applied Spectroscopy, vol. 46, No. 4, (1992) pp. 553-559.|
|60||Turner et al., "LCTF Raman Chemical Imaging in the Near-Infrared," SPIE vol. 3061 (1997) pp. 280-283.|
|61||Turner et al., "Near-Infrared Acousto-Optic Tunable Filter Hadamard Transform Spectroscopy," Applied Spectroscopy, vol. 50, No. 2, (1996) pp. 227-284.|
|62||Zhang et al., "Details of Damage Profile in Self-Ion-Implanted Silicon," vol. 25 Journal of Raman Spectroscopy, (1994) pp. 515-520.|
|Citing Patent||Filing date||Publication date||Applicant||Title|
|US8054454||27 Aug 2008||8 Nov 2011||Chemimage Corporation||Time and space resolved standoff hyperspectral IED explosives LIDAR detector|
|US8368880||5 Apr 2010||5 Feb 2013||Chemimage Corporation||Chemical imaging explosives (CHIMED) optical sensor using SWIR|
|US8582089||6 Oct 2010||12 Nov 2013||Chemimage Corporation||System and method for combined raman, SWIR and LIBS detection|
|US8729502||21 Oct 2011||20 May 2014||The Research Foundation For The State University Of New York||Simultaneous, single-detector fluorescence detection of multiple analytes with frequency-specific lock-in detection|
|US9103714||6 Oct 2010||11 Aug 2015||Chemimage Corporation||System and methods for explosives detection using SWIR|
|US20060124443 *||18 Oct 2005||15 Jun 2006||David Tuschel||Control and monitoring of non-resonant radiation-induced nucleation, crystallization, and polymorph formation|
|US20100225899 *||5 Apr 2010||9 Sep 2010||Chemimage Corporation||Chemical Imaging Explosives (CHIMED) Optical Sensor using SWIR|
|US20110080577 *||6 Oct 2010||7 Apr 2011||Chemlmage Corporation||System and Method for Combined Raman, SWIR and LIBS Detection|
|US20110089323 *||6 Oct 2010||21 Apr 2011||Chemlmage Corporation||System and methods for explosives detection using SWIR|
|US20110237446 *||7 Apr 2011||29 Sep 2011||Chemlmage Corporation||Detection of Pathogenic Microorganisms Using Fused Raman, SWIR and LIBS Sensor Data|
|US20120147224 *||22 Nov 2011||14 Jun 2012||Canon Kabushiki Kaisha||Imaging apparatus|
|U.S. Classification||356/73, 356/417, 250/339.02, 356/318, 250/339.12, 356/51, 250/339.09|
|International Classification||G01J3/28, G02B21/06, G01N21/35|
|Cooperative Classification||G02B21/06, G01N21/359, G02B21/0016, G01J3/2823|
|European Classification||G01N21/35G, G02B21/06, G01J3/28D|
|2 Nov 2011||FPAY||Fee payment|
Year of fee payment: 8
|27 Aug 2012||AS||Assignment|
Owner name: CHEMIMAGE CORPORATION, PENNSYLVANIA
Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNORS:TREADO, PATRICK J.;NELSON, MATTHEW;KEITZER, SCOTT;SIGNING DATES FROM 20040112 TO 20040126;REEL/FRAME:028856/0619
|9 Jun 2013||AS||Assignment|
Owner name: CHEMIMAGE TECHNOLOGIES LLC, PENNSYLVANIA
Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNOR:CHEMIMAGE CORPORATION;REEL/FRAME:030573/0454
Effective date: 20130530
|11 Nov 2015||FPAY||Fee payment|
Year of fee payment: 12